[Lab] Zing 24 analysis
henri.kuschkowitz at gmail.com
Tue Sep 1 09:49:15 EDT 2015
Thanks for leading the way and fixing it.
On September 1, 2015 at 9:48:15 AM, Stephen Burke (steve at envirolaser.com) wrote:
According to the tech support at Epilog, and based upon all the testing/servicing they had me do, there are two components that would be most likely the cause of the problem.
sensor board (75%)
sensor cable (20%)
The sensor cable connects the sensor board to a control board. There is only a 5% chance that it is the control board. Of course, you have to change the sensor board and cable first to determine if it is the control board.
I have submitted a quote to Artengine; if accepted the parts will be ordered ASAP and the unit could be fully operational by September 25th. Again, this all depends on the diagnosis being correct.
Lab mailing list
1. subscribe http://artengine.ca/mailman/listinfo/lab
2. then email Lab at artengine.ca to send your message to the list
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the Lab