[Lab] Zing 24 analysis
steve at envirolaser.com
Tue Sep 1 09:47:56 EDT 2015
According to the tech support at Epilog, and based upon all the testing/servicing they had me do, there are two components that would be most likely the cause of the problem.
sensor board (75%)
sensor cable (20%)
The sensor cable connects the sensor board to a control board. There is only a 5% chance that it is the control board. Of course, you have to change the sensor board and cable first to determine if it is the control board.
I have submitted a quote to Artengine; if accepted the parts will be ordered ASAP and the unit could be fully operational by September 25th. Again, this all depends on the diagnosis being correct.
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the Lab